You don’t need anything from Agilent and the find instrument drivers is not supposed to work. Then click on the RUN button again as this is a single measurement. Let’s look at the Block Diagram for the function generator. This block diagram also demonstrates the complexity of the advanced designs used to control test instruments. The NI Instrument Driver Network houses thousands of instrument drivers for automating third-party instruments. Using the mouse draw the cursor around all the items in the design. Note, this feature is only supported on the E power supply.
|Date Added:||14 March 2012|
|File Size:||61.80 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 10 of How do I initialize the port? You can find these instrument drivers in the Instrument Driver Network.
I don’t think that the problem lies there. By keeping the File Path input to labvied function blank, the program prompts the user to browse to a location on disk and give the file a name.
Agilent Technologies / Keysight Technologies hpe363xa Power Supply
We were only able to operate one Agjlent at a time. Press that and observe how your design behaves. The model for the new function generator is the Agilent A. Sequencing and Database Reporting With TestStand Test Management Software TestStand is a ready-to-run test management software is designed to help you develop, execute, and deploy automated test and validation systems faster.
You can download the parser and its associated SDK from: Have all the bench instruments turned off at the start. Message 8 of If you install the. However, I get the very same error when I plug my power supply into the serial port of my computer and try to run the test VIs.
EA – Want display to show voltage/current i | Keysight Community
There are also default com port settings in the Device Manager of Windows. This driver supports instruments manufactured by Agilent Technologies. The resulting Block Diagram is shown in Figure 6 below. Are you saying by 363a that you are able to send and receive commands when connected to a “built in” comm port of your PC using a program like Terra Term or Hyper Terminal, but not able to send and receive commands when using a 3rd party “RS to Agulent converter?
E3631A – Want display to show voltage/current in realtime
Navigate labvisw the ‘Hardware Interfaces’ palette to find the ‘Configure Tracking’ function Connect the purple instrument wire and yellow error wires through the function to the Read Output. How often does your dual instrument VI update the data? I selected my port from the dropdown menu in the front panel and then started the VI.
Then click the green Run arrow at the top left of the VI. You have to be a little careful in selecting the correct GPIB address, so before turning on the multi-meter you should take a look at your resources currently 3361a. Note that the oscilloscope has not been configured as a VI and is controlled by the actual front panel push buttons.
Wentworth Institute of Technology. The very first function you need to use is the initialize. So I got myself a serial-usb-adapter and downloaded the driver from the NI website: Another student is looking at the display and starts counting the flashes for the duration of the timing test. Wire the actual voltage level, voltage level, actual current limit, and actual current limit to the four instances of the Number to Fractional String function.
SystemDesigner displays a high-level view of your system.
It is also available from: That seems like the kind of error you get when you’re trying to connect to a device that isn’t recognized or is on the wrong port. For the old function generator use the Agilent 33XXX series folder to get to those examples.